Your 3M™ Littmann® Stethoscope is designed to provide years of reliable service. By routinely cleaning your stethoscope, you can help extend its life and acoustic performance.
Practicing regular stethoscope care and using your stethoscope correctly as outlined in your stethoscope’s user manual will not only help prolong its continued acoustic performance, but also aid in the upkeep of its tubing colour and chestpiece finish. Most importantly, it will contribute to the overall maintenance of hygiene at your workplace.
Tunable diaphragms, non-chill bell sleeves and eartips can be removed for thorough cleaning with 70% alcohol or mild soapy water. Ensure that all parts and surfaces are dried thoroughly before reassembly.
|Non-chill bell sleeve||
Some stethoscope models have warranty periods that go up to 7 years. Registering for a warranty online will only take about a minute. Be sure to have your serial number ready.
You can clean your 3M™ Littmann® Stethoscope between patient exams, and to remove organic material. The practice of using hand sanitizers to clean stethoscopes is not recommended because of additives that may damage parts of the stethoscope, such as the diaphragms.
We recommend these cleaning solutions for all 3M™ Littmann® Stethoscope models:
✓ Mild soap and water
✓ 70% alcohol
✓ 2% bleach solution (may cause tubing discolouration)
✕ Use hand sanitizers
✕ Immerse stethoscope in any liquid
✕ Subject your stethoscope to any sterilisation process
Never rinse or soak your 3M™ Littmann®
Stethoscope. Clean it with an alcohol wipe,
or a wipe dipped in an appropriate
Proper storage of your stethoscope is important to increase its lifespan and prevent damage to its functional and aesthetic aspects. There are many ways of storing your stethoscope for extended periods of time or in harsh environments – here are some of them.
Where to store your stethoscope:
• In the box that it came in.
• In a natural position inside a cool and dry cupboard, drawer or locker.
• Do not expose your stethoscope to extreme heat or cold.
• Store your stethoscope away from direct sunlight as continuous exposure will harden its tubing.
• Do not place/keep your stethoscope near solvents and oils.
• Do not place heavy objects on your stethoscope as it could excessively warp or bend.
• Do not store it tightly folded in pockets.
Your stethoscope will be one of your most-used instruments at work, which makes it susceptible to wear and tear after some time. Here are some common indicators that it may be time to repair your 3M™ Littmann® Stethoscope, find replacement parts, or invest in a new stethoscope.
1. Its parts are worn and torn, or weather-beaten from sunlight exposure.
2. Its tubing becomes hard or cracked due to being exposed for long periods of time to skin oils.
3. Loss of functionality due to e.g. immersion in water
4. Damage to the diaphragm(s), eartips, and non-chill bell sleeves.